a. Backrest, leg rest, auto contour, height (up - down), Trendelenburg and reverse Trendelenburg positions are controlled by 4 DC motors.
b. Positions are controlled by “Hand Remote Control”.
c. Angle indicator on each side rail of the bed.
d. Head and foot boards are made of PP plastic and easily clearable. These parts can also be assembled and disassembled easily.
e. 4 pieces of restraint strap holder (2 pieces on each side).
f. Lying surface consists of 4 pieces easily clearable ABS plastic.
g. 4 pieces folding side rails are made of PP plastic and have release system assisted by gas spring.
h. Automatic back sliding and leg sliding functions are available in the bed.
j. Metal chassis is epoxy polyester electrostatic painted and covered with ABS plastic.
k. Bumpers are located at each corner of the bed to protect the bed against crushes.
l. Accessory sockets are located at each corner of the bed for IV poles or traction sets.
m. One adjustable IV pole is provided with the bed.
n. There are drainage bag holders at both sides of the bed.
o. Mattress is waterproof and zippered.
p. Removable Polypropylene (pp) plastic head and foot ends.
r. Patient Control Locking & Automatic Contour: